# Electromagnetic reverberation chamber

{{ safesubst:#invoke:Unsubst||$N=Use dmy dates |date=__DATE__ |$B= }}

An **electromagnetic reverberation chamber** (also known as a **reverb chamber (RVC)** or **mode-stirred chamber (MSC)**) is an environment for electromagnetic compatibility (EMC) testing and other electromagnetic investigations. Electromagnetic reverberation chambers have been introduced first by H.A. Mendes in 1968.^{[1]} A reverberation chamber is screened room with a minimum of absorption of electromagnetic energy. Due to the low absorption very high field strength can be achieved with moderate input power. A reverberation chamber is a cavity resonator with a high Q factor. Thus, the spatial distribution of the electrical and magnetic field strength is strongly inhomogeneous (standing waves). To reduce this inhomogeneity, one or more *tuners* (*stirrers*) are used. A tuner is a construction with large metallic reflectors that can be moved to different orientations in order to achieve different boundary conditions. The *Lowest Usable Frequency* (LUF) of a reverberation chamber depends on the size of the chamber and the design of the tuner. Small chambers have a higher LUF than large chambers.

The concept of a reverberation chambers is comparable to a microwave oven.

## Glossary/Notation

### Preface

The notation is mainly the same as in the IEC standard 61000-4-21.^{[2]} For statistic quantities like mean and maximal values, a more explicit notation is used in order to emphasize the used domain. Here, *spatial domain* (subscript ) means that quantities are taken for different chamber positions, and *ensemble domain* (subscript ) refers to different boundary or excitation conditions (e.g. tuner positions).

### General

- : Vector of the electric field.
- : Vector of the magnetic field.
- : The total electrical or magnetical field strength, i.e. the magnitude of the field vector.
- : Field strength (magnitude) of one rectangular component of the electrical or magnetical field vector.
- : Characteristic impedance of the free space
- : Efficiency of the transmitting antenna
- : Efficiency of the receiving antenna
- : Power of the forward and backward running waves.
- : The quality factor.

### Statistics

- : spatial mean of for objects (positions in space).
- : ensemble mean of for objects (boundaries, i.e. tuner positions).
- : equivalent to . Thist is the expected value in statistics.
- : spatial maximum of for objects (positions in space).
- : ensemble maximum of for objects (boundaries, i.e. tuner positions).
- : equivalent to .
- : max to mean ratio in the spatial domain.
- : max to mean ratio in the ensemble domain.

## Theory

### Cavity resonator

A reverberation chamber is cavity resonator—usually a screened room—that is operated in the overmoded region. To understand what that means we have to investigate cavity resonators briefly.

For rectangular cavities, the resonance frequencies (or eigenfrequencies, or natural frequencies) are given by

where is the speed of light, , and are the cavity's length, width and height, and , , are non-negative integers (at most one of those can be zero).

With that equation, the number of modes with an eigenfrequency less than a given limit , , can be counted. This results in a stepwise function. In principle, two modes—a transversal electric mode and a transversal magnetic mode —exist for each eigenfrequency.

The fields at the chamber position are given by

Due to the boundary conditions for the E- and H field, some modes do not exist. The restrictions are:^{[3]}

- For TM modes: m and n can not be zero, p can be zero
- For TE modes: m or n can be zero (but not both can be zero), p can not be zero

A smooth approximation of , , is given by

The leading term is proportional to the chamber volume and to the third power of the frequency. This term is identical to Weyl's formula.

Based on the *mode density* is given by

An important quantity is the number of modes in a certain frequency interval , , that is given by

### Quality factor

The Quality Factor (or Q Factor) is an important quantity for all resonant systems. Generally, the Q factor is defined by where the maximum and the average are taken over one cycle, and is the angular frequency.

The factor Q of the TE and TM modes can be calculated from the fields. The stored energy is given by

The loss occurs in the metallic walls. If the wall's electrical conductivity is and its permeability is , the surface resistance is

where is the skin depth of the wall material.

The losses are calculated according to

For a rectangular cavity follows^{[4]}

- for TE modes:

- for TM modes:

Using the Q values of the individual modes, an averaged *Composite Quality Factor* can be derived:^{[5]}

includes only losses due to the finite conductivity of the chamber walls and is therefore an upper limit. Other losses are dielectric losses e.g. in antenna support structures, losses due to wall coatings, and leakage losses. For the lower frequency range the dominant loss is due to the antenna used to couple energy to the room (transmitting antenna, Tx) and to monitor the fields in the chamber (receiving antenna, Rx). This antenna loss is given by where is the number of antenna in the chamber.

The quality factor including all losses is the harmonic sum of the factors for all single loss processes:

Resulting from the finite quality factor the eigenmodes are broaden in frequency, i.e. a mode can be excited even if the operating frequency does not exactly match the eigenfrequency. Therefore, more eigenmodes are exited for a given frequency at the same time.

The *Q-bandwidth* is a measure of the frequency bandwidth over which the modes in a reverberation chamber are
correlated. The of a reverberation chamber can be calculated using the following:

Using the formula the number of modes excited within results to

Related to the chamber quality factor is the *chamber time constant* by

That is the time constant of the *free energy relaxation* of the chamber's field (exponential decay) if the input power is switched off.

## See also

## Notes

- ↑ Mendes, H.A.:
*A new approach to electromagnetic field-strength measurements in shielded enclosures.*, Wescon Tech. Papers, Los Angeles, CA., August, 1968. - ↑ IEC 61000-4-21:
*Electromagnetic compatibility (EMC) - Part 4-21: Testing and measurement techniques - Reverberation chamber test methods*, Ed. 2.0, January, 2011. ([1]) - ↑ Cheng, D.K.:
*Field and Wave Electromagnetics*, Addison-Wesley Publishing Company Inc., Edition 2, 1998. ISBN 0-201-52820-7 - ↑ Chang, K.:
*Handbook of Microwave and Optical Components*, Volume 1, John Willey & Sons Inc., 1989. ISBN 0-471-61366-5. - ↑ Liu, B.H., Chang, D.C., Ma, M.T.:
*Eigenmodes and the Composite Quality Factor of a Reverberating Chamber*, NBS Technical Note 1066, National Bureau of Standards, Boulder, CO., August 1983.

## References

- Crawford, M.L.; Koepke, G.H.:
*Design, Evaluation, and Use of a Reverberation Chamber for Performing Electromagnetic Susceptibility/Vulnerability Measurements*, NBS Technical Note 1092, National Bureau od Standards, Boulder, CO, April, 1986. - Ladbury, J.M.; Koepke, G.H.:
*Reverberation chamber relationships: corrections and improvements or three wrongs can (almost) make a right*, Electromagnetic Compatibility, 1999 IEEE International Symposium on, Volume 1, 1-6, 2–6 August 1999.